|Email Newsletters from Pharmaceutical Technology and Pharmaceutical Technology Europe|
News from Europe's pharmaceutical manufacturing industry coupled with upcoming events, and exclusive articles and interviews from industry experts.
Verification Methods for 198 Common Raw Materials Using a Handheld Raman Spectrometer
Raman spectroscopy's inherent selectivity makes it an effective option for qualitative verification of pharmaceutical raw materials (1–3). Until recently, Raman spectroscopy was difficult to implement because of the technical challenge of detecting the relatively weak but highly selective Raman scattering effect. During the past 10 years, improvements in the sensitivity of charge-coupled device (CCD) detectors, the increased availability of longer-wavelength diode lasers, and developments in Rayleigh rejection filters have made Raman spectroscopy suitable for routine applications. Moreover, miniaturization of optical components has enabled the development of portable, handheld Raman spectrometers capable of robust material verification through transparent packaging materials such as glass and plastic (4).
US Pharmacopoeia General Chapter ‹1120› recognizes Raman spectroscopy as one of the accepted technologies for material identity verification, asserting that "Because the Raman spectrum is specific for a given compound, qualitative Raman measurements may be used as a compendial ID test, as well as for structural elucidation" (5). As this statement suggests, the inherent characteristics of Raman spectra, namely discrete peaks corresponding to specific molecular-bond vibrations, enable verification of a wide range of raw materials.
In addition to its high degree of molecular selectivity, Raman spectroscopy is relatively insensitive to physical and environmental factors such as particle size, humidity, and contributions from packaging material. This insensitivity combined with the high molecular selectivity allows for very streamlined method development when using Raman. In fact, a method can be typically based on a single reference spectrum (4).
The high molecular selectivity and external factor insensitivity of Raman spectroscopy help eliminate problems associated with transferring methods between multiple instruments. Careful instrument engineering to provide a consistent spectroscopic platform to acquire and share data, with thorough system response characterization, allow a method developed on one device to be shared seamlessly with any other device, thereby saving time and cost (4).
Portable, field-based alternatives offer many operational advantages over laboratory testing, but it is important to consider challenges arising when such a technology is in use such as the inability to control the testing environment (temperature, humidity, ambient lighting, etc.). Instruments used in the field should be expressly designed for field use and be able to automatically compensate for external variables such as ambient lighting and sampling geometry, and spectroscopic differences in analyzed materials.
An important final consideration when evaluating any analytical methodology is the method performance that can be achieved. In the context of identity testing, specificity (i.e., selectivity) is the only figure of merit that must be demonstrated through comprehensive validation (6). When making an identity assessment on the basis of spectral data, the unknown measurement is examined in relation to one or more reference spectra. The final assessment can be performed manually by comparing peak locations, but usually a full-spectrum automated analysis is performed. Clearly, the algorithmic approach used to render a final decision plays a role in determining the level of specificity that can be achieved when performing an analysis. Common approaches for spectral comparison include hit quality index (HQI) and spectral correlation (7). These methods are useful for quick similarity assessments; unfortunately, they generally do not provide a direct interpretation in the context of spectral identity testing. Furthermore, even when these techniques produce a favorable correlation coefficient, misidentification may occur because these metrics are not particularly sensitive to discrepancies between spectra of interest (4).
An alternative to HQI and spectral correlation is to use a fully multivariate probabilistic approach that looks for spectroscopic discrepancies between the test and reference spectrum in question. Under a null hypothesis that the test and reference spectra are the same within the measurement uncertainty, the comparison is distilled into a traditional p-value (small p-values indicate greater discrepancy between test and reference spectra). If no significant discrepancies are found (i.e., p-value >0.05), then the device deems the test spectrum consistent with the reference spectrum, and a pass result is declared. If discrepancies are found—other than those expected to arise from the uncertainty in the measurement alone—then the test material cannot be authenticated against the reference and a fail result is declared.
In the present study, the broad applicability of portable Raman spectroscopy for raw material identity verification is evaluated by examining a range of 198 Raman-active substances widely used as raw materials in the manufacture of pharmaceutical, consumer health, and personal care products. Each verification method was developed using one reference spectrum in a fraction of the time typically required for near-infrared (NIR) method development. Many diverse categories of materials were tested (see sidebar, "Categories of materials"). A thorough assessment of selectivity was achieved by evaluating each material against all methods to fully assess the performance of each method.
Acquiring a spectrum is facilitated by using an auto-acquisition mode that monitors incoming data in real time and automatically adjusts the control settings (e.g., laser power, exposure time, number of exposures) such that spectra of comparable statistical resolving power are achieved from one measurement to the next. This adaptive approach to spectroscopy ensures consistent statistical performance of the instrument and embedded decision algorithms regardless of operating conditions.
Four of the instruments (hereafter called reference devices) were used to acquire reference spectra for the 198 materials included in this study (see sidebar, "Materials tested"). To obtain the reference spectra, a reference sample of each material contained in a borosilicate glass vial (Kimble Glass, Vineland, NJ) was placed at the correct distance from the device's laser aperture using a vial-holder accessory. All data acquisition was performed in auto-mode. For test spectra, most analyses are complete in less than one minute. Reference-grade spectra used for method development are also generated using an auto-mode; however, these spectra have higher standards for signal-to-noise ratio than a standard test spectrum and therefore require a longer data acquisition time. Most auto-mode reference spectra are acquired in less than 30 min. The scans that did not auto-terminate within 45 min were manually stopped at 45 min to prevent photobleaching.
Reference spectra can be transferred between TruScan devices using a compact flash card or a standard web browser that addresses the device via a TCP/IP connection. Using these approaches, the reference spectra were transferred from the four reference devices to the test device. Once all of the reference spectra had been transferred to the test device, the TCP/IP utility WebAdmin was used to generate a new method for each of the 198 materials using only one reference spectrum for each method. In the context of the TruScan device, methods are analytical tasks based on stored reference spectra that the instrument's software executes to determine whether a material's identity can be verified.
To challenge the methods, a test sample of each material was placed in a borosilicate glass vial, and the vial was inserted into the test device's vial holder. For each test sample, a spectrum was acquired using auto-mode. The onboard processor then evaluated the spectrum acquired for each test material against the method developed for that material, and also against all the methods created for the rest of the 198 materials, for a total of 39,204 evaluations of test spectra against methods. In addition to the pass and fail results normally presented by the test device, the p-value was recorded for each test sample–method evaluation pair.
Results and discussion
The handheld Raman material-identity verification system shows a pass result whenever test of a method against a material produces a p-value ≥0.05. In general, testing a method against the "wrong" material produces a p-value far less than 0.05, resulting in a fail.
The false-positive results for the six pairings shown in Figure 3 are results of the extremely high similarity in composition and/or molecular structure of the materials in each pair. One example is safflower oil, which has a very high concentration of linoleic aicd. Other examples are povidone and crospovidone which have the same structure except for a cross linkage in crospovidone. The slight differences between the materials in each pair do not result in significant differences in their Raman spectra.
To enhance the granularity of p-value results, the range was more finely divided in the following compound class studies, allowing a more detailed evaluation of the differentiating capability of the device's embedded decision algorithms.
1. T. Vankeirsbilck et al., "Applications of Raman Spectroscopy in Pharmaceutical Analysis," TrAC 21 (12), 869–877, (2002).
2. R.L. McCreery et al., "Noninvasive Identification of Materials Inside USP Vials with Raman Spectroscopy and a Raman Spectral Library," J. Pharm. Sci. 87 (1), 1–8 (1998).
3. R. Cantu et al., "A Simple Approach for Developing, Validating, and Transferring an Identification Method for Multidose Pharmaceutical Products using FT-Raman Spectroscopy," Am. Pharm. Review 10 (2), 96–103, 111 (2007).
4. R.L. Green and C.D. Brown, "Raw-Material Authentication Using a Handheld Raman Spectrometer," Pharm. Technol. 32 (3), 148–163 (2008).
5. USP 32–NF 27 ‹1120› "Raman spectrometry," p. 1120.
6. ICH Q2R1 Harmonised Tripartite Guideline, Validation of Analytical Procedures: Text and Methodolgy (Geneva, Switzerland, 2005).
7. S.R. Lowry, "Automated Spectral Searching in Infrared, Raman, and Near-Infrared Spectroscopy," in Handbook of Vibrational Spectroscopy, Vol.3 (Wiley, New York, 2001), pp. 1948-1961.
Robert L. Green* is a research scientist, Robert Brush is an application scientist, Wayne Jalenak is a senior application
scientist, and Christopher D. Brown is director of system analytics and applications, all at Ahura Scientific, 46 Jonspin
Rd., Wilmington, MA 01887, tel. 978.657.5555, fax 978.657.5921, firstname.lastname@example.org
*To whom all correspondence should be addressed.
Submitted: Jan. 6, 2009. Accepted: Feb. 3, 2009.