OR WAIT 15 SECS
Volume 22, Issue 7
Wyatt Earp, the legendary sheriff of Tombstone used to solve troubles in a simple way: aiming, pulling the trigger and bang! ... problem eliminated! Some manufacturers check the readability of their barcodes in the same way.
The full version of this counterfeiting feature can be read in the July issue of our digital magazine: http://www.pharmtech.com/ptedigital0710
Wyatt Earp, the legendary sheriff of Tombstone used to solve troubles in a simple way: aiming, pulling the trigger and bang! ... problem eliminated! Some manufacturers check the readability of their barcodes in the same way: pointing, pulling the trigger and beep! However, do they really solve the issue? Unfortunately, no.
Before looking at the details, let us ask why barcode quality is a hot topic for the pharma industry. The primary reason is because of the mass serialisation of folding boxes, which has been mandated by several health authorities including France and Turkey, based on Data Matrix barcode technology. Therefore, maintaining an acceptable level of barcode quality in pharmaceutical production processes will be an increasingly important task, and a legal requirement.
The error-free readability of the code should be guaranteed throughout the global supply chain. The barcode issuer is responsible for the barcode's readability by all readers, including older and even worn out readers. Just using a normal handheld reader to check the readability of the code is not sufficient because its reading quality is not defined. Instead, specialised testing equipment and measuring procedures are required, and a full range of ISO/IEC standards have been created to define barcode quality levels (grades) and measuring systems (verifiers), as well as the related testing methods.
Jason Hosking/Getty Images
In this article, I will show why assessing barcode quality is important and will report on some of my own experiences from the practice.
The fight against counterfeiting and the need to assure patient health are the main drivers for deploying mass serialisation on pharmaceutical products — applying a unique code to each package dramatically increases the transparency of the supply chain and makes criminal activities easier to detect. Because of the advantages, mass serialisation is being promoted by many health authorities, including those in Brazil, China, India and Spain; however the main drivers of implementation at the moment are the following:
Verifiers are required to measure the readability of barcodes. Such an instrument is normally based on a digital camera, with specific illumination and software to process the image data — the relevant design criteria and measuring tolerances of the verifier are specified in the ISO/IEC standard 15426 (parts 1 and 2).5,6 As most barcode readers use red illumination, the verifiers also deploy red light of typically 620 nm to 640 nm wavelength. This is an important restriction because it is impossible to print barcodes in red, yellow, orange or brown as these colors are poorly visible in red light. Black, blue and green tones are recommended for printing barcodes.
ISO/IEC standards define the measuring procedures for validating barcode quality: ISO/IEC 154167 sets rules for linear barcodes and ISO/IEC 154158 for Data Matrix codes. To ensure that all instruments give the same absolute values, a global calibration standard is important. For linear barcodes, such a standard has been available for many years, but until recently a standard was missing for Data Matrix codes. Fortunately GS1, the issuing agency of the EAN 13 linear barcode, which is in daily usage on almost all products, has now developed a "Calibrated Conformance Standard Test Chart" for Data Matrix barcodes.9 This calibration chart features seven codes with specific defects to check the correct reading of all measuring parameters, thus assuring absolute correct levels for all numerical code quality data generated by the verifier.
Barcode quality is measured in grades that can be reported as numbers from 0 to 4, or as letters from A to F (Figure 1). Grade A(4) is the best value and grade F(0) is the worst, indicating that the code is not readable ("F" stands for Failure).
Very often, a barcode may, at first glance, seem so perfect that even experts would consider it a highgrade code. A good example is shown in Figure 2; however, the code is not readable and is graded F by the verifier. Why? A macroscopic investigation shows the problem. The code is geometrically distorted, is not symmetrical and therefore does not fit into the reference grid. The lesson learned is straightforward: the human eye cannot replace a specialised measuring instrument.
The ISO/IEC standard 15415 mandates that the sample shall be measured five times and rotated under the camera by 72° between each measurement,8 but there have been numerous discussions about how practically relevant this requirement is, especially as before 2009 there were no instruments on the market that allowed for a comfortable standard ISO measuring cycle. At the end of 2009, however, Compar (Switzerland) teamed up with Cognex (MA, USA) to launch a new product designed to specifically fulfill the ISO/IEC 15415 Data Matrix verification requirement.
The system enables a clear correlation between barcode grades and sample rotation to be observed by testing low quality Data Matrix barcodes. Measures of codes with geometrical distortions, such as the one in Figure 2, are especially sensitively to rotation — the value of the grades can vary as much as between A and F for the same code depending on the rotation angle! Data Matrix barcodes properly printed with a measured grade B or better, on the other hand, are practically insensitive to rotation. Figure 3 shows the measuring results of good and poor codes.
Most Data Matrix barcodes with variable data content are printed in-line on automated packaging lines. Because continuous monitoring of the barcode quality is highly beneficial, a camera system can be connected to a powerful computer to run the quality assessment calculation at high cadence. Until recently, the available computing power allowed only the use of a low end grading procedure based on the ISO/IEC 16022 specification for automatic identification and data capture techniques.10 Unfortunately, the results of the inline corresponding validation cannot be compared with the officially accepted data as described above because different parameters are used for specifying the grades.
Thanks to increasingly powerful computers, however, the ISO/IEC 15415 measuring procedure can now also be used for in-line quality monitoring. Because the inline quality check is based on the same computing algorithm as the standalone verifier, direct comparisons can be made; however, we have to keep in mind that the inline system will hardly provide the same accuracy because of limitations, such as very short measuring times, illumination restrictions, vibrations and dust. Additionally, the in-line system can only take one picture, while offline verifier computes a grading based on five pictures taken at five different angles. Nevertheless, experience shows a good correlation of the data generated by the in-line camera and offline verifier as long as the barcodes are of impeccable quality. For low grade codes, however, the comparability is not as good because rotation is necessary to assure accurate quality measurements.
As already mentioned, using the ISO/IEC 15415 standard requires calibrating the measuring instrument and, consequently, this obligation also applies for in-line measuring equipment. A reference chart carrying a pattern or code with a well-defined contrast is required for the calibration process.
Data Matrix barcodes used for serialising products are, in most cases, called 'concatenated codes' because they carry structured information consisting of one or more data strings, with every one prefixed by an Application Identifier (AI). This AI is a numerical header positioned in front of each data string in the code. The AIs are globally standardised to guarantee proper decoding of the data content. Additionally, some technical characters must be present in the code to allow the reader to identify the type of code and the encoding scheme — these characters can be considered as an operating system for the code, called syntax. If the syntax and structure of a code are not both perfect, the data content of the code cannot be interpreted. Therefore, it is highly recommended that the syntax and structure of the code are also checked during the verification process.
You can't control what you can't measure, and this wisdom also applies to barcode technology.
Assessing the quality of barcodes is not only a legal requirement, but is mandatory from engineering and quality assurance perspectives to assure proper system function. In fact, more and more end dealers and distributors, including Walmart, are fining suppliers when they deliver products carrying low grade barcodes!11
The most effective method of supervising the Data Matrix barcode quality level of packaging lines is to implement an inline monitoring camera system based on the ISO/IEC 15415 norm. Because of accuracy limitations, inline systems cannot replace standalone verifiers, but when used together the systems complement one another; the inline system allows screening mass production while the standalone verifier can be used as an in-process control instrument to validate rejected samples, as well as to service and maintain the packaging line.
1. EUR-Lex http://eur-lex.europa.eu
2. Standards for Securing the Drug Supply Chain - Standardized Numerical Identification for Prescription Drug Packages (FDA, June 2009). www.fda.gov
3. Notice to human-use medicinal product marketing authorization holders and head pharmacists of the pharmaceutical establishments cited in article R. 51242 of the French Public Health Code (CSP) (Afssaps, Paris, France, February 2007). www.afssaps.fr
4. Guidance on Implementation of Identification and Barcoding of Medicinal Products for Human Use Version 1.1 (T.C. Turkish Ministry of Health, G.D. of Pharmaceuticals and Pharmacies, Ankara, Turkey, 2008). www.iegm.gov.tr
5. ISO/IEC 15426-1:2006 Information technology – Automatic identification and data capture techniques – Bar code verifier conformance specification – Part 1: Linear symbols (ISO, April 2007). www.iso.org
6. ISO/IEC 15426-2:2005 Information technology – Automatic identification and data capture techniques – Bar code verifier conformance specification – Part 2: Two-dimensional symbols (ISO, June 2009).
7. ISO/IEC 15416:2000 Information technology – Automatic identification and data capture techniques – Bar code print quality test specification – Linear symbols (ISO, April 2008).
8. ISO/IEC 15415:2004 Information technology – Automatic identification and data capture techniques – Bar code print quality test specification – Two-dimensional symbols (ISO, June, 2009).
9. GS1 DataMatrix: An introduction and technical overview of the most advanced GS1 Application Identifiers compliant symbology (GS1, 2009). www.gs1.org
10. ISO/IEC 16022:2006 Information technology – Automatic identification and data capture techniques – Data Matrix bar code symbology specification (ISO, June 2009).
11. OCR Canada www.ocr.ca/barcode/barcode.asp