OR WAIT 15 SECS
To fully optimize MS performance, corresponding enhancements are required in the other functional components of the integrated LC/MS platform.
Mass spectrometry (MS) is seen by many as a primary detection technology in drug discovery and development as well as in pharmaceutical QA/QC. Molecular mass is arguably the most broadly discriminative analytical parameter. Conjoined with LC retention time, it provides a distinctive signature for each analyte across the broad sweep of molecular entities. In addition, the ability to measure relative abundance also makes MS a first-order tool for quantification. Despite its large and long-standing contributions, MS technology has not fully met modern pharma application needs.
In target discovery, for the selectivity necessary to discern individual molecular signatures, mass accuracy is key. For picking up low-abundance biomarkers indicative of a pathway or metabolic response an ultra-low level of detection (LOD) is desired. This has only been available from very high priced MS.
In drug discovery, the focus has traditionally been on speed and throughput. A brute force approach to the workflow has made it possible to generate and then rapidly evaluate large numbers of molecular candidates for potency against multiple screens. Over time, however, this approach has yielded diminishing returns and increasing operational costs.
For QA/QC, with its massive and increasing processing load, throughput will always be issue number one. Add to that the need for better sensitivity, robust and repeatable performance, and ease-of-use.
New MS solutions to the challenges described above are now emerging. They are derived from key advances such as:
These advances have given rise to new levels of performance, ease-of-use and throughput for many applications. In some cases, they are facilitating entirely new work flows.
Examples of such application shifts include pre-empting time consuming biomarker identification analyses with profiling-based methodologies in an effort to reduce the number of required IDs to be preformed. Such proteomic and metabolomic profiling techniques are the direct result of new high mass accuracy LC/MS TOF instrumentation that can give uniqueness to literally thousands of analytes in a single analysis. Mass accuracies in the 1–2 ppm level that extend over a wide analyte concentration are the result of sophisticated calibration and reference mass correction algorithms that have been realized via simple software control. When the application centres on the small or large molecule confirmation, the same features result in a completely automated "walk up analysis", wherein a user places samples in the instrument's autosampler, selects the method, initiates the run command, and in due course receives the results in a preselected report format.
Multimode ionization source technology has improved many screening techniques as well as applications in QA/QC. Multimode ionization gives greater universality by simultaneously generating ions by both ESI and APCI processes under a very broad range of separation flow-rate conditions. These advances eliminate the need for replicate analysis, resulting in higher throughput.
To address the need for cost-effective performance, ruggedness and ease-of-use, there are new generation instruments such as the triple quadrupole MS with femtogram-level sensitivity. Intended for a variety of pharma analytical applications, these machines are exceptionally reliable and are supported by easy-to-learn operating and end user customizable reporting software. Innovations of this sort are particularly welcome in quantification of drug candidates from early in vitro studies through clinical trials where the requirement for high throughput remains unabated and where challenges such as sample preparation, formulation complexity and detection limits are becoming more demanding.
To fully optimize MS performance, corresponding enhancements are required in the other functional components of the integrated LC/MS platform. Here too, there are exciting developments pushing the performance envelope. New rapid-resolution HPLC systems capable of order-of-magnitude increases in throughput have recently been introduced. On another technology front, the microfluidic HPLC chip is revolutionizing column chromatography. The chips are designed to integrate with and maximize the entire continuum of LC/MS performance. Benefits include easier operation, greater application flexibility, increased uptime and lower operational costs.
Ultrasensitive and diverse MS technology, high throughput, rapid-resolution LC, microfluidic chips and new informatics — these are the key components of emerging instrument platforms that will support the growing suite of analytical applications across the pharma enterprise.
Taia Ergueta is general manager for proteomics, metabolomics and LC/MS at Agilent Technologies, USA.